In this article, you will learn where are the best locations of temperature data loggers placement during validations thermal mapping, and other environmental monitoring in Puerto Rico. Moreover, here you can discover how to get a thermal mapping validation template and validation management software application to handle your qualification activities fast and easily.
Once you estimate and calculate how many data loggers or sensors should be necessary for your thermal mapping, the next step is to determine where they shall be placed and installed to start the temperature monitoring.
Normally, all data loggers shall be distributed throughout each location in which your product shall be placed to get a detailed understanding of the thermal profile.
Moreover, it is important to identify the most relevant cold and hot spots within the area and determine locations:
1. To place your product.
2. To place your permanent monitoring probe(s)
3. Establish the alarm setup configuration.
It is highly recommended to install a data logger very close and adjacent to the monitoring/alarm probe/s. The information of this data logger can be used to correlate and confirm that the alarm probe location is the most suitable for the monitoring probe i.e. the highest or lowest temperature within the area.
You can install your monitoring probe in one of the following locations depending on the purpose of your permanent monitoring system:
1. Location closest to the average temperature within the area. This is the location least likely to fail since any fault in the area is unlikely to be detected until a prolonged period of time or the next thermal mapping exercise.
2. Location closest to the highest heat location (called hot spot location). This is the location most likely to fail in the high-temperature limit since any fault with the system going too hot is very likely to be detected.
Location closest to the lowest to heat location (called cold spot location).
This is the location most likely to fail in the low-temperature limit since any fault with the system going too cold is very likely to be detected.
In order to have a balanced system of monitoring and alarm, some people use two monitoring and alarm probes placed within the area. These two (2) data loggers or probes could be located in the minimum and the maximum temperature locations identified during the temperature mapping. Indeed, these two probes shall be able to identify any faults with high and low temperatures in the area.
In case that your control system is very sensitive and frequently generate many false alarms due to temporary excursions as part of the normal behavior of the area, it is highly recommended to install the monitoring probes with some kind of retardant, buffers (Glycol, aluminum block well, etc.) or use an alarm delay to prevent false alarms caused by routine use (door opening, etc.)
CIQA can bring you validation services, as well as the rental of the data loggers to make the temperature monitoring.
Thermal studies require the use of very expensive validation instrumentation and equipment, which represent a high cost in many cases if the utilization of that equipment is very low.
Since some companies do not have the internal personnel with the experience and knowledge to perform these studies, they contract third-party companies as CIQA Engineering to make that work. Thermal Mapping Outcomes
CIQA Engineering has the equipment and experienced professionals to perform the thermal mapping and qualifications. We can prepare the protocols, bring the measurement instruments for monitoring, make the analysis and reporting documents once the data is gathered.
Recommendations to Avoid Mistakes Selecting Temperature Data Loggers.
We suggest evaluating all your requirements and needs to use temperature data loggers, but, also consider the frequency of use monthly, and how fast you need them on-site for your validation study. Thermal Mapping Distribution Studies
So, you can determine if is better to buy or rent them based on the expertise and time availability of your employees to configure and prepare the trending graphs and summary statistical reports on time. Thermal Mapping Outcomes
Before buy, consider the cost of the data loggers, but, also the cost of the calibration and its lead time.
Evaluate and compare the total costs and lead times, against the costs to rent data loggers already calibrated.
If you are NOT using temperature data loggers frequently, consider a fast CIQA rental of the calibrated data loggers required for the thermal study. The
VALIDATION PROTOCOL TEMPLATES
GET THERMAL MAPPING AND VALIDATION PROTOCOL TEMPLATES
In case you need to create or update your thermal mapping protocol, you can use our template protocol to accelerate the process to perform the validation study and monitoring.
Now you can download a complete validation protocol template fully editable in MS Word or get a hard-copy and Amazon Kindle format, ready to fill, and use according to your cGMP needs, check the following documents and see more information at the CIQA Shop.
DATA LOGGERS SETUP AND CONFIGURATION SERVICES:
One of our data logger experts can set up, configure and bring to you the data loggers ready to use and be installed to collect information from your thermal mapping study. At the end of your monitoring, you can send the data loggers back to us for data extraction, evaluation, and perform the statistical analysis and plotting with averages, maximums, and minimums.
cGMP VALIDATION & REGULATORY COMPLIANCE CONSULTING
VALIDATION MANAGEMENT SOFTWARE APPLICATION
MORE SERIES AND VIDEOS ABOUT TEMPERATURE DATA LOGGERS
TO RECEIVE NEWS AND USEFUL TOOLS ABOUT THE DIFFERENCES BETWEEN TEMPERATURE DATA LOGGERS, SUBSCRIBE, AND FOLLOW US.
CIQA is a quality and regulatory consultant with 25 years of experience developing products and managing projects in the medical device supply chain, and pharmaceutical industries. Rest assured of the safety or being in the right hands.
- What is a Temperature Data Logger?
- The 12 Most Common Applications for Temperature Data Loggers.
- The Top 6 Advantages Using Temperature Data Loggers
- The 5 Most Common Types of Temperature Data Loggers
- The Top 10 Main Characteristics of the Temperature Data loggers
- The 4 Main Attributes of Temperature Data loggers.
- Costs of the Temperature Data Loggers for Validation Purposes
- The 4 Best Features of the Temperature Data Loggers
- The Top 3 Pros and Cons of Temperature Dataloggers
- The 5 Most Common Uses of the Temperature Data loggers
- How to Determine the Data Loggers Record Duration?
- How to Download the Information out of the Temperature Data Logger
- How to Determine the Number of Temperature Data Loggers Do I Need?
- How Data loggers can handle their memory.
- The Top 6 Reasons Why People Rent Temperature Data Loggers.
- 3 Reasons Why are Temperature Data Loggers Useful.
- How to Configure a temperature Data logger in 5 steps
- The 5 Most Common Mistakes When Buying Data Loggers
- How to Choose the Best Temperature Data Loggers for your Validation.
- 6 Main Differences between Data Loggers.
- What is a Thermal Mapping Distribution Study?
- How to Make a Thermal Mapping in Four Steps
- What 5 Tests Should Be Performed During Temperature Mapping?
- Understanding the 5 Most Relevant Thermal Mapping Outcomes
- The Best Locations to Place Temperature Data loggers for thermal mappings.
https://youtu.be/H58OKuwg6Cc – Ep27 software
https://youtu.be/YxrzxgN0R5o – Ep26 template
https://youtu.be/MKrmP3EGhG4 -Ep6 – old
https://youtu.be/J82Q3H9c0IE – Ep10 Differences between data loggers
https://youtu.be/PuEfgrOdm4I – Ep11
https://youtu.be/oeaWhumdi5A – Ep12
https://youtu.be/3p7Q8jHkAUA – Ep13
https://youtu.be/GDf_pB4opbE – EP14